The AccuSync 72 ECG trigger monitor for gated cardiac studies automatically detects the R Wave with precision and reliability. Its menu-driven Color Active Matrix Display combines ease of use with increased functionality and flexibility while its light and compact design makes it portable.
UL2601-1/CSA601.1 Certified by TUV Rheinland.
Less than 10 μA with 230 V~, 60 Hz applied to the patient leads. Patient related circuitry is isolated from ground by more than 4000 V, 5500 V peak.
5000 V, 10 ms duration.
90db with 51 kOhm imbalance.
Diagnostic: 0.05 to 100Hz
LCD TFT Display: 0.5 to 50Hz
Printer: 0.5 to 100Hz
3 lead system, RA, LA and LL with 6 pin connector.
5 lead system; RA, LA, RL, LL and C with 6 pin connector.
50 nA DC
100 μV peak
The R trigger is a 5V TTL/CMOS (or 12V) compatible square wave for the computer trigger; independent of ECG signal polarity. R Wave detection and trigger to computer is indicated by audio signal and by presence of a superimposed dot on the R Wave which appears on display. The output can deliver 35 mA of current or drive up to 150 feet long RGU 174/U cable. R trigger is compatible with all computers including Digirad, GE, Hitachi, MDS, Philips, Siemens and Toshiba.
RS232 interface available to transmit and receive information via an IBM or compatible computer. In transmit mode, the interface will send the ECG data to the computer while in receiver mode the computer will control the parameters of the instrument.
ECG Output: Variable gain analog ECG signal up to x2000 available on a BNC connector
DIMENSIONS: Height 6.0? Width 9.0? Depth 7.0?
POWER: 100-240 V~ 60/50 Hz .30/.15 A
Display Area: Height=98mm Width=132.48mm
Display indicates: heart rate, R-R interval, ECG size, ECG lead selection, audio, freeze and print indicator, R trigger event marker.
Menu driven selection of parameters displayed on 16 color TFT Active Matrix Display
Paper Speed: 25mm per second. When FREEZE is selected, recorder prints last 8 sec of data.
Printing Method: thermal line dot system.
Resolution: Horizontal-8 dots per mm. Vertical-16 lines per mm.
Event Marker: When R Wave is detected, a mark is superimposed on the R Wave to indicate the presence of an R trigger signal triggering the computer.
Nickel Metal Hydride (NiMH) 12V
Allows user to delay R trigger output up to 1000 ms. Result appears on TFT display.